Replacing damaged or missing screws Before working on the open MULTIMOBIL 2.5 switch off the unit with the power switch on the control panel and unplug the power cable. Before inserting or removing PC boards, observe the ESD regulations.
Legal regulations may contain special prescriptions concerning the disposal of this product. In order to avoid environmental damage and/or personal injury, please inform SIEMENS Customer Service if you want to put the unit out of operation and dispose it. 8.4.1 Radioprotection Material Lead in X-ray tube assembly and in Collimator 1.65 kg.
Check function of locking lever and cap screw on the X-ray tube. Check the legibility of the labels according to those in section Location of Labels. If any label requires to be replaced, please notify Siemens customer service. Performance Check (Every Six months) Check Counterbalancing of the support arm at each position.
8.10 Removing the top panel , front cover and X-ray source assembly 8.10.1 Removing the top panel Turn off the Multimobil 2.5,disconnect the mains chord and wind the cord around the cable holder. Lower the X-ray source assembly to transport position. Lock the Multimobil 2.5 by applying the brake.
Position the front cover on a soft bedding. 8.10.3 Removing the X-ray source assembly. Turn off the Multimobil 2.5,disconnect the mains chord and wind the cord around the cable holder. Lower the X-ray source assembly to transport position. Lock the movement of the Multimobil 2.5 by applying the foot brake.