Fieldbus Foundation. OUNDATION IrDA is a registered trademark of the Infrared Data Association. The Emerson logo is a trademark and service mark of Emerson Electric Co. All other marks are the property of their respective owners. NTRODUCTION The 375 Field Communicator can make your work easier and more productive by ®...
Product overview and precautions RODUCT OVERVIEW AND PRECAUTIONS The 375 Field Communicator supports HART and F fieldbus devices, letting OUNDATION you configure or troubleshoot in the field. When using the 375 Field Communicator to communicate with devices, follow all standards and procedures applicable to the location.
Product overview and precautions • The battery must not be charged in hazardous area environments. • Use the 375 Field Communicator power supply/charger (00375-0003-0005) with only the 375 Field Communicator. • Use only the power supply/charger to charge the battery. Failure to comply may permanently damage your 375 Field Communicator and will void the IS approval and the warranty.
Assembly SSEMBLY Before operating the 375 Field Communicator, ensure: • The 375 Field Communicator is not damaged. • The battery is fully seated. • All screws are sufficiently tightened. • An Expansion Module or Expansion Port Plug is in place. •...
Installing the System Card and battery Figure 3. Back side of the 375 Field Communicator Main unit label Stand assembly Battery retaining IS label screws (KL Option) System Card Battery NSTALLING THE YSTEM ARD AND BATTERY 1. Place the 375 Field Communicator face down on a level, secure surface. 2.
Starting the 375 Field Communicator 375 F TARTING THE IELD OMMUNICATOR 1. On the keypad, press and hold the On/Off key until the multifunction LED blinks (approximately two seconds). During start up, the 375 Field Communicator checks the System Card for any software upgrades and notifies you if an upgrade is required.
The label indicates this product should be recycled and not treated as household waste. Customers in EU member states should contact their Emerson sales representative for information on discarding any part of the 375 Field Communicator. For customers in all other world areas, if it is necessary to discard any part of the 375 Field Communicator, adhere to the waste-disposal regulations applicable in your locality.
Tested to specifications EN 61000-6-3, EN 61000-6-2, and EN 61326-1. ATEX Directive (94/9/EC) (KL option only) Emerson Process Management complies with the ATEX Directive. Specific ATEX Directive Information is located within this document and the 375 Field Communicator User’s Manual.
Product certifications European certifications ATEX Intrinsic Safety Certification No.: BVS 03 ATEX E 347 II 2 G (1 GD) Ex ia IIC T4 (-10 °C ≤ Ta ≤ +50 °C) 1180 HART Intrinsically Safe electrical parameters Input Parameters = 30 Volt DC = 200 mA = 1.0 Watt Output Parameters...