ACUSON X300 ACUSON X300 PE ACUSON X150 Supplement to the User and Reference Manuals S i e m e n s M e d i c a l S o l u t i o n s U S A, I n c .
Printed in the Republic of Korea. Directive 93/42/EEC of June 14, 1993 concerning Medical Devices. The CE marking only applies to ACUSON is a trademark of Siemens Medical Solutions medical devices that have been put on the market USA, Inc.
Contents Supplement to the User and Reference Manuals..........4 About this Supplement .................. 4 Compatible Systems and Product Versions ........... 4 Wireless Network Connections ................. 5 Approved Wireless Adapters ................. 5 Connecting the Wireless Adapter to the System ........... 6 Informational Icons for the Wireless Adapter..........
This supplement provides new information about features, accessories, settings, and standards. Refer to the user and reference manuals for a complete set of operating instructions for your ultrasound system. Compatible Systems and Product Versions ACUSON X300 PE (7.5) ACUSON X300 (2.5) SONOVISTA X300 (2.5, 7.5) ...
5 GHz frequency bandwidth to reduce potential radio interference from the 2.4 GHz frequency bandwidth. Approved Wireless Adapters Caution: Use of wireless adapters not tested for use and approved by Siemens can damage the ultrasound system. Do not connect any wireless adapters to the system except the wireless adapters listed below.
RF energy sources. To prevent this interference, increase the distance between the ultrasound system and the interfering RF energy source. Siemens recommends connecting the wireless adapter to the leftmost USB port on the Input/Output panel of the ultrasound system. Informational Icons for the Wireless Adapter...
Wireless Network Selections (System Configuration) (Supplement to System Configuration, Chapter 1, System Reference) Use the system presets to enable and disable a wireless network adapter connected to the ultrasound system. You can require a password to prevent unauthorized changes to the wireless network selections.
Network Profile Settings Wireless > Network Settings > New Wireless > Network Settings > Edit Selection Option(s) Allows you to... Network Name (text entry) Enter the name of a wireless network. (SSID) Non-broadcast on, off Configure the network profile for a wireless network that network broadcasts its name (off) or does not broadcast its name (on).
Internet Protocol Properties Wireless > Connection Information > Properties Selection Option(s) Allows you to... Obtain an IP address automatically (DHCP) on, off Configure the ultrasound system to use dynamic IP addresses assigned by a DHCP (Dynamic Host Configuration Protocol) server (on) or to use a static IP (Internet Protocol) address (off).
Access the system presets. Select Wireless. Enter a password, if required. Do This: Activate a connection to a wireless Note: Siemens recommends using secured networks to ensure data security. network Select Enabled. Deactivate the connection to a Select Disabled.
Ultrasound System Security (Virus Protection) Ultrasound System Security is virus protection software that Siemens recommends for ultrasound systems connected to a LAN or wireless network. To enable Ultrasound System Security, contact your Siemens service representative.
1. Press Report. The system displays the patient report using the layout provided by Qview. 2. Press Report to exit the report page. Or, for the ACUSON X300 PE system and the ACUSON X150 system, press the Exit soft key.
Report Option — Configuration Selections Report Option Selection Option(s) Allows you to... Report Type System Report Only Select the layout the ultrasound system uses to display Telexy (Qview) Only patient reports. Both System Report Only enables only the layout provided by the ultrasound system.
QuickStart Battery-Powered Standby Mode (Available only on the ACUSON X300 PE 7.5 system) (Supplement to QuickStart Feature (Battery-Powered Standby Mode), System Setup, Chapter 3, Instructions for Use) The QuickStart battery pack charges whenever the ultrasound system is connected to a power outlet, including when the system is powered on, powered off, or in standby mode.
Do This Display or hide the Press the toggle switch on the handle of the transducer. reference line indicating the intersection of the sagittal and transverse planes on each image Reposition the sagittal Press the STEER control on the control panel. image during live dual format Example of live dual format before the sagittal image is repositioned.
Needle Guide Brackets (Supplement to Safety and Care, Chapter 2, Instructions for Use and Transducer Accessories and Biopsy, Chapter 5, Instructions for Use) BP9-4 Disposable Endocavity Biopsy Straight Needle Guide Kit 1 The disposable endocavity needle guide is a single-use item. Refer to the in-box instructions for attachment and care procedures, including cleaning, sterilization, and disposal.
Tissue Harmonic Imaging (THI) Frequencies (Available only on the ACUSON X300 PE 7.5 system) (Replaces the frequency settings for only the following three transducers for THI, Imaging Functions, Technical Description, Appendix A, Instructions for Use; the frequencies for other compatible transducers are unchanged) ...
Standards Compliance (For ACUSON X300 PE 7.5: replaces the Standards Compliance, Technical Description, Appendix A, Instructions for Use; the Acoustic Output Standards and CE Declaration are unchanged.) (For ACUSON X300 2.5: replaces the Standards Compliance, Technical Description, Chapter 6,  Instructions for Use; the Acoustic Output Standards and CE Declaration are unchanged.)