QX and SX recorders - Installation Instruction
This Installation Instruction sheet is intended as a guide for replacing or installing hardware and for setting up
functionality in the recorder. Refer to the User manual for detailed operational requirements.
The QX and SX recorders are designed for ease of assembly with minimal disturbance to the rest of the unit. If unsure,
please return the unit to your supplier for repair or upgrade.
Removing the case and opening the back of the recorder should only be performed under the following circumstances:
When an item of hardware requires individual replacement.
When an item of hardware is to be retrospectively fitted.
In all other instances it is recommended that the complete unit be returned to an authorised agent or service centre.
For Agency Approved recorders the product needs to be upgraded or repaired by an Authorized Repair facility
Disconnect all power to the recorder before removing the case and attempting any maintenance procedures.
Upon completion of service procedures detailed in this manual two basic safety tests should be performed in order to
ensure continued safe operation of the instrument.
Earth Resistance; 25 VDC applied between case and protective earth, bonding resistance should be < 0.1 Ohm.
Insulation Resistance; 500Vdc applied between the earth terminal, and the live and neutral terminal shorted together,
insulation resistance to be >2.0 MOhm. (Mega Ohm, NOT milli Ohm)
Failure to comply with these instructions could result in death or serious injury
OBSERVE ANTI-STATIC PRECAUTIONS
Refer to BS EN61340-5-1: 2001. Basic specification. Protection of electrostatic sensitive devices.
Full anti-static precautions MUST be observed when in contact with the electronics of your recorder.
SAVE DATA, SETUPS AND LAYOUTS
Removal of PCBs and battery back-up will result in the loss of all non-volatile data.
Ensure all data and set-ups are saved.
Failure to comply with these instructions may result in product damage.
43-TV-33-59 iss.2 GLO Jan 21 UK