PRODUCT DESCRIPTION The Vive Precision blood pressure monitor is a portable device that allows you to accurately track blood pressure from the comfort of home. It is rigorously tested for accuracy and approved for use by the FDA, and has a built in alarm function to alert you of irregular heartbeats.
POWER SUPPLY The Choice of Power Supply 1. Battery powered mode: 6VDC 4×AAA batteries 2. AC adaptor powered mode: (Please use the recommended AC adaptor model). (Not Included) AC adaptor *Please unplug the adaptor when not in use. REPLACING BATTERIES WARNING In order to get the best effect and protect your monitor, please use the right battery and special power adapter which...
MEASUREMENT PRINCIPLE This product uses the Oscillometric Measuring method to detect blood pressure. Before every measurement, the unit establishes a “zero pressure” equivalent to the air pressure. Then it starts inflating the arm cuff, meanwhile, the unit detects pressure oscillations generated by beat-to-beat pulsatile, which is used to determine the systolic and diastolic pressure, and also pulse rate.
3. When you get the right year, press “S” button to set down and turn to next step. 4. Repeat step 2 and 3 to set the [MONTH] and [DAY]. 5. Repeat steps 2 and 3 to set the [TIME FORMAT between 12 hour time and 24 hour time.
7. Repeat step 2 and 3 to set the [MEASUREMENT UNIT]. mmHg is U.S. Standard Measurement 8. After the unit is set, the LCD will display “done” first, then display all the settings you have done and then it will turn off. OPERATING INSTRUCTIONS - APPLYING THE CUFF 1.
4. Sit comfortably with your test arm resting on a flat surface. 5. Patients with Hypertension: The middle of the cuff should be at the level of the right atrium of the heart; Before starting measurement, please sit comfortably with legs uncrossed, feet flat on the floor, back and arm supported.
LCD Display Adjust the zero. Display and save the results. Inflating and measuring. 2. Press the “START/STOP” to power off, otherwise it will turn off within 1 minute. Tips: Combined maximum 250 records for User A and User B.
OPERATING INSTRUCTIONS - DATA MANAGEMENT Recall the Records 1. When the monitor is off, press the “M” button to show the average value of the latest three records for the selected user. (Take User 1 for example. See right.) Note: It will display the latest record first when the records are less than three groups.
WARNING The most recent record (1) is shown first. Each new measurement is assigned to the first (1) record. All other records are pushed back one digit (e.g., 2 becomes 3, and so on), and the last record (250) is dropped from the list. Delete the Records If you did not get the correct measurement, you can delete each user A &...
INFORMATION FOR USER 4. Immediate measurement after tea, coffee, or smoking. 5. When talking or moving your fingers. 6. When you need to urinate. Maintenance In order to get the best performance, please follow the instructions below. WARNING If you have any problems with this device, such as setting up, maintaining or using, please contact firstname.lastname@example.org.
ABOUT BLOOD PRESSURE CONTINUED This chart reflects blood pressure categories defined by American Heart Association. Systolic Blood Pressure Diastolic mmHg (upper#) Category mmHg (lower#) less than 120 Normal less than 80 Prehypertension 120-139 80-89 140-159 High Blood Pressure 90-99 (Hypertension) Stage 1 High Blood Pressure (Hypertension)
ABOUT BLOOD PRESSURE CONTINUED we recommend you seek medical advice. Please note that the device does not replace a cardiac examination, but serves to detect pulse irregularities at an early stage. Why does my blood pressure fluctuate throughout the day? 1.
TROUBLESHOOTING This section includes a list of error messages and frequently asked questions for problems you may encounter with your blood pressure monitor. If the products not operating as you think it should, check here before arranging for servicing. PROBLEM SYMPTOM CHECK THIS REMEDY...
AUTHORIZED COMPONENT Use the Vive® authorized adaptor. (Not Included) Adaptor Type: UE08WCP-060100SPA Input: 100~240V, 50~60Hz,400mA Output: 6V (Conforms to UL certificate) COMPILED STANDARDS LIST ISO/EN 14971:2012 Medical devices — Application of risk Risk management management to medical devices Labeling ISO/EN 15223-1:2012 Medical devices. Symbols to be used with medical device labels, labelling and information to be supplied.
SPECIFICATIONS • Power Supply: Battery Powered Mode: 6V DC 4 x AAA Batteries AC Adaptor Powered Mode: 6V 1A (Please only use recommended AC adaptor - not included.) • Display Mode: Digital LCD V.A. 60mm x 92mm • Measurement Mode: Oscillographic testing mode •...
SAFETY INFORMATION Symbol for “THE OPERATION GUIDE MUST BE READ” Symbol for “MANUFACTURER” Symbol for “SERIAL NUMBER” Symbol for “DIRECT CURRENT” Symbol for “MANUFACTURE DATE” Symbol for “TYPE BF APPLIED PARTS” Symbol for “ENVIRONMENT PROTECTION-Waste electrical products should not be disposed of with household waste.
WARNING • The equipment is not AP/APG equipment and not suitable for use in the presence of a flammable anesthetic mixture • with air of with oxygen or nitrous oxide. • The operator shall not touch output of batteries and the patient simultaneously.
WARNING • Do not apply the cuff over a wound, otherwise it can cause further injury. • Do not inflate the cuff on the same limb which other monitoring ME EQUIPMENT is applied around simultaneously, because this could cause temporary loss of function of those simultaneously-used monitoring ME EQUIPMENT.
2 W yields d=3, 3m at an IMMUNITY LEVEL of 3V/m) GUARANTEE REGISTRATION You are protected by Vive Health’s industry leading guarantees and customer service: 2 YEAR GUARANTEE If you did not purchase through vivehealth.com, please register at vivehealth.
We sincerely appreciate your business. We strive to provide you with the best quality products at great value. If you have any questions please contact us. email@example.com 1-800-487-3808 Monday - Friday 9am - 5pm EST...