1 Introduction .................1 1-1 Conventions Used in This Manual........... 1 1-2 Computer Requirements ............1 1-3 Notes on using AG-150A with the integrated centralized control software (TG-2000A)..............1 2 Setting the Operating Environment........2 2-1 Setting the PC IP Address............2 2-2 Setting the Web Browser............
Pack." Note: Use a security device such as a VPN router when connecting the AG-150A to the Internet to prevent unauthorized access. Note: "Air to water booster unit" and "Air to water HEX unit" are referred to as "Hot water supply".
2-1 Setting the PC IP Address You need to set an IP address on the PC that will enable you to connect to the AG-150A using a web browser. For instance, if the AG-150A IP address is [192.168.1.1], the PC IP address will need to belong to the same system (for example [192.168.1.101]).
(4) In the [Internet Protocol (TCP/IP) Properties] dialog, click [Use the following IP address] and enter the IP address (for example, “192.168.1.101”) that you want to set in the IP address field. You normally set [255.255.255.0] as the subnet mask. Note: Ask your LAN administrator to provide the IP addresses and subnet mask.
2-2 Setting the Web Browser Perform the necessary web browser settings to enable the web browser to connect to the AG-150A. Note: The settings and screen images used as examples in this manual are based on Internet Explorer 6.0. 2-2-1 Not connecting to the Internet If the PC you use for monitoring air conditioners and performing operations is not going to be connected to the Internet, use the procedure given below to set the web browser environment settings.
Note: If connecting to more than one AG-150A, you can specify multiple IP addresses like [192.168.1.1; 192.168.1.2], however, it is also possible to use the asterisk (*) and specify [192.168.1*].
AG-150A. Note: If the AG-150A is restarted due to circumstances like a power interruption, wait until the screen on the AG-150A main unit displays the normal operation screen (it takes several minutes before the normal operation screen is displayed) before using a web browser to access the AG-150A.
AG-150A can encrypt communication data using HTTPS (SSL). When connecting the AG-150A to the LAN that can be accessed by the general public, it is recommended that the following settings be made so that the units are monitored and controlled on the encrypted Web page.
(4) If the security alert has not been disabled, after entering the web address and hitting the Enter (Return) key, a security alert message will alert asking if you want to proceed . This is because the AG-150A uses the self-authentication system.
(7) Enter the web page address in the address field of the web browser as follows and press the [Enter] key on the keyboard. A screen appears for optional (measurement) settings. https://[IP address of the AG-150A]/g-50/option.html Note: For example, type “https://192.168.1.1/g-50/option.html” if the AG-150A IP address is [192.168.1.1].
AI controller (PAC-YG63MCA) and PI controller (PAC-YG60MCA). After the settings are made, click [Save Settings] to send setting data to AG-150A. Note: License of "Web Monitor" or "Basic License Pack" is necessary to use the web browser and update program. Register the license key on the AG-150A LCD screen.
3-2-3 Setting the trend data format The data format to be used when the trend graph data of the measured value is downloaded can be set. (1) Select the separator character between [Comma ( , )] and [Semicolon ( ; )]. (2) Select the decimal point character between [Period ( .
[Enter] key on the keyboard. A screen appears for login. http://[IP address of the AG-150A]/administrator.html Note: For example, type “http://192.168.1.1/administrator.html” if the AG-150A IP address is [192.168.1.1]. Note: To encrypt data, change [http] to [https].
3-3-3 Using a graph Click [Trend Graph] on the measurement condition list screen, and select the measured item to be displayed. The measured value can be checked by graph. The graph can also be downloaded as a CSV file. Overview display Click to display the measurement Update to most recent value as a list.
Item Description Click [Download] to download the graph being displayed. The data format of the downloaded data is as follows. Item Format First line File type 124: Outdoor unit data yyyy/mm/dd Second line Date Note: The date format set on Initial Setting Web.will be applied. Third line Trend target Temperature: "Address"...
HEAD OFFICE: MITSUBISHI DENKI BLDG., 2-2-3, MARUNOUCHI, CHIYODA-KU, TOKYO 100-8310, JAPAN WT05493X01...