TABLE OF CONTENTS TABLE Of CONTENTS Before using the Monitor Introduction ... 3 Safety Information ... 4 Operating the Device ... 4 Care and Maintenance ... 5 Operating Instructions Before Taking a Measurement ... 6 Know Your Unit ... 7 Unit Display ...
The monitor is compact and easy-to-use at home, at work, and during travel. Perfect for people who frequently monitor their own blood pressure. Your new digital blood pressure monitor uses the oscillometric method of blood pres- sure measurement. This means the monitor detects your blood’s movement through your brachial artery and converts the movements into a digital reading.
SAfETY INfORMATION To assure the correct use of the product, basic safety measures should always be followed including the warnings and cautions listed in this instruction manual. SAFETY SYMBOLS USED IN THIS INSTRUCTION MANUAL Indicates a potentially hazardous situation which, if not avoided, WARNING could result in death or serious injury.
SAfETY INfORMATION Do not use a cellular phone near the device. It may result in an operational failure. Use only Omron authorized parts and accessories. Parts and accessories not approved for use with the device may damage the unit. Use only 1.5 V “AAA” alkaline batteries with this device. Do not use other types of batteries.
BEfORE TAKING A MEASUREMENT To ensure a reliable reading follow these recommendations: 1. Avoid eating, drinking alcohol, smoking, exercising, and bathing for 30 minutes before taking a measurement. Rest for at least 15 minutes before taking the measurement. 2. Stress raises blood pressure. Avoid taking measurements during stressful times. 3.
Fill in for future reference. DATE PURCHASED: __________________ SERIAL NUMBER: ____________________ • Staple your purchase receipt here Instruction Manual Wrist cuff Memory (MEM) button START/STOP button User ID selection switch SET button 5329415-9A ® BP652 IntelliSense ® Automatic Wrist Blood Quick Start Guide...
UNIT dISPLAY Positioning indicator Diastolic blood pressure Memory symbol (Displayed when viewing values stored in memory) Systolic blood pressure Average value symbol (Refer to “Using the Deflation symbol Memory Function” .) Irregular heartbeat symbol Movement error symbol (Displayed if you move Blood pressure level your body during the indicator...
IRREGULAR HEARTBEAT SYMBOL ( When the monitor detects an irregular rhythm two or more times during the measurement, the Irregular Heartbeat Symbol ( measurement values. An irregular heartbeat rhythm is defined as a rhythm that varies by less than 25% from the average rhythm or 25% more than the average rhythm detected while the monitor is measuring the systolic and diastolic blood pressure.
dISPLAY SYMBOLS This is a general guideline because blood pressure depends on age and morbidity. A lower home blood pressure goal may be recommended for certain patients, including diabetic patients, pregnant women, and patients with renal failure. WARNING Contact your physician for specific information about your blood pressure. Self-diagnosis and treatment using measured results may be dangerous. Follow the instructions of your physician or licensed healthcare provider.
BATTERY INSTALLATION CAUTION Use only 1.5V “AAA” alkaline batteries with this device. 1. Press the hook on the bottom of the battery cover and lift the cover off in the direction of the arrow. 2. Install 2 “AAA” size batteries so the + (positive) and - (negative) polarities match the polarities of the battery compartment as indicated. 3. Set the date and time. Refer to “Setting the Date and Time”.
BATTERY INSTALLATION BATTERY REPLACEMENT When the Low Battery Indicator appears on the display, turn the monitor off and remove all the batteries. Replace with 2 new batteries at the same time. Long-life alkaline batteries are recommended. WARNING If battery fluid should get in your eyes, immediately rinse with plenty of clean water. Contact a physician immediately. NOTES: • Turn the unit off before replacing the batteries. If the batteries are removed while the unit is still on, the date and time will be reset to that of the previous use.
SETTING THE dATE ANd TIME Set the monitor to the current date and time before taking a measurement for the first time. 1. TO START THE SETTING Press the Date/Time Setting button ( ) to adjust the date and time. The year flashes on the display. 2. SETTING THE YEAR Press the Date/Time Setting button ( ) to advance by increments of one year.
SETTING THE dATE ANd TIME 4. SETTING THE DAY Press the Date/Time Setting button ( ) to advance by increments of one day. Press the Date/Time Setting button ( ) to set the current day. The hour flashes on the display. 5. SETTING THE HOUR The time is set using AM or PM.
SETTING THE AdVANCEd POSITIONING SENSOR (APS) You can change the options for various settings on your monitor. Pressing the SET button to change a setting. Press the MEM button to select the options for that setting. After selecting a setting, press the START/STOP button to confirm the setting and turn the power off. SETTING THE POSITIONING INDICATOR 1.
SETTING THE AdVANCEd POSITIONING SENSOR (APS) SETTING FOR RIGHT OR LEFT WRIST MEASUREMENT 1. While in the setting mode, press the SET button until the wrist symbol ( NOTES: • The default setting is “ ”. • After you choose “oFF” for the Positioning indicator, the monitor skips this process. 2. Press the MEM button to select the left wrist ( ) or the right wrist ( ). SETTING THE BUZZER 1.
SETTING THE AdVANCEd POSITIONING SENSOR (APS) If the Positioning indicator is “on1” or “on2”: on1: The buzzer sounds only when the monitor is in proper position. on2: The buzzer sounds a series of two short blips if your wrist is too far away from the measuring position, and beeps when your wrist is in the proper position. oFF: The buzzer will not sound.
APPLYING THE WRIST CUff CAUTION Read all of the information in the instruction manual and any other literature in the box before operating the unit. CAUTION This device is intended for use in measuring blood pressure and pulse rate in the adult population. Do not use this device on infants or persons who cannot express their intentions.
APPLYING THE WRIST CUff 5. Fold the remaining part of the wrist cuff back out of the way. APPLYING THE CUFF ON THE RIGHT WRIST When taking a measurement using the right wrist position the cuff as shown in the illustration. Refer to “Setting for Right or Left Wrist for Measurement”.
TAKING A MEASUREMENT The monitor is designed to take measurements and store the measurement values in memory for two people using USER ID A and USER ID B. The monitor can be used to take a single measurement using the GUEST mode are not stored in the memory. 1. Sit comfortably in a chair with your feet flat on the floor. 2. Select your User ID (A or B). 3.
TAKING A MEASUREMENT 5. Press the START/STOP button. All symbols appear on the display. The cuff starts to inflate automatically. As the cuff inflates, the monitor automatically determines your ideal inflation level. This monitor detects your blood pressure and pulse rate during inflation. The Heartbeat Symbol ( ) flashes at every heartbeat. Remain still and do not move until the entire measurement process is completed. NOTE: To stop the inflation or measurement, push the START/STOP button. The monitor will stop inflating, start deflating, and will turn off. 5. After the monitor has detected your blood pressure and pulse rate, the cuff automatically deflates. Your blood pressure and pulse rate are displayed.
USING THE MEMORY fUNCTION Every time you complete a measurement the monitor stores the blood pressure and pulse rate for each user (A and B) in the memory. The monitor automatically stores up to 100 sets of measurement values (blood pressure and pulse rate) for each user (A and B).
USING THE MEMORY fUNCTION NOTES: • The date and time are alternately displayed with the measurement values. • Press the MEM button repeatedly to display the next values. • Press and hold the MEM button to display the values faster. 3. Press the START/STOP button to turn the monitor off. TO DELETE ALL VALUES STORED IN THE MEMORY The values stored in the memory are deleted by User ID. You cannot partially delete values stored in the memory.
CARE ANd MAINTENANCE To keep your digital blood pressure monitor in the best condition and protect the unit from damage, follow the directions listed below: Do not forcefully bend the cuff. Do not fold tightly. Clean the monitor with a soft dry cloth.
ERROR INdICATORS SYMBOL Monitor could not detect pulse wave. Cuff under-inflated. Cuff not applied correctly. Cuff over-inflated. Device error. Movement during measurement. Irregular or weak pulses are detected. Batteries are worn. CAUSE Take another measurement and remain still until the measurement is complete.
TROUBLESHOOTING TIPS PROBLEM No power. No display appears on the unit. Measurement values appear too high or too low. CAUSES AND SOLUTIONS Replace both batteries with new ones at the same time. Check the battery installation for proper placement of the battery polarities. Blood pressure varies constantly.
fCC STATEMENT Note: POTENTIAL FOR RADIO/TELEVISION INTERFERENCE (for U.S.A. only) This product has been tested and found to comply with the limits for a Class B digital device, pursuant to part 15 of the FCC rules. These limits are designed to provide reasonable protection against harmful interference in a residential installation.
LIMITEd WARRANTY Your BP652 IntelliSense Automatic Wrist Blood Pressure Monitor, excluding ® the wrist cuff and batteries, is warranted to be free from defects in materials and workmanship appearing within 5 years from the date of purchase, when used in accordance with the instructions provided with the monitor.
Memory: Contents: NOTE: These specifications are subject to change without notice. SPECIfICATIONS BP652 (HEM-6052-Z) LCD Digital Display Pressure: 0 to 299 mmHg, Pulse: 40 to 180/min. Pressure: ±3mmHg or 2% of reading Pulse: ±5% of reading Automatic by electric pump...