Model 1134 AccuRead Blood Pressure Monitor & Clock User Manual Distributed by: The Lumiscope Co., Inc www.lumiscope.net Repair Department: 1-800-672-8293 Made and Printed In China...
You can store up to 99 measurements in each of the unit's 4 memory zones, ideal for users who want to monitor and track their blood pressure on a regular basis. The 1134 is compact and portable, making it perfect for home use and travel.
How the 1134 Works Your model 1134 uses the oscillometric method to detect your blood pressure. Before the cuff starts inflating, the device will establish a baseline cuff pressure, which is equivalent to the air pressure. The measurement of your blood pressure is based on this baseline pressure.
Parts Identification Start/Power Memory Zone Select Average Selection Memory Select WHO Indicator Unit Display Symbols Display Symbols Pre-formed Cuff Air Tube USB Plug /PC Link Cuff Plug Outlet AC Adaptor Outlet Date/ Time Systolic Pressure Diastolic Pressure Pulse Value...
• Operated by 4 "AA" batteries or AC/DC adapter. Note: This unit also offers PC-LINK software. If you order this function, you will receive a separate package including a CD and a connection line with a USB plug. You may order this accessory at our website www.lumiscope.net.
Installing the Batteries 1. To insert the batteries A. Press down on the latch and lift the cover towards you to open the battery compartment. B. Insert 4 "AA" batteries as indicated. Make sure to match polarities (+, –) on the batteries. C.
Applying the Cuff To apply the cuff A. Plug the end of the tubing on the cuff into the connector at the base of the monitor. B. Remove any clothing that restricts the upper left arm. Wrap the cuff around your upper left arm. The rubber tube should point in the direction of your hand.
Measurement Procedures To start the measurement Note: In order to get your correct measurement result and to avoid erroneous measurements you should be seated quietly, your arm, hand, and fingers should be still during measurement procedure. A. Push the "PAGE" button to select one of the four Memory Zones.
Irregular Heartbeat Dectection An irregular heartbeat may cause or indicate cardiovascular disease. This unit has a unique function that alerts the user to the presence of an irregular heartbeat. Irregular Heartbeat reading When taking your blood pressure reading, If you are experiencing an irregular heartbeat the unit will display the Pulse Value and "IH"...
Memory Recall To see the previously measured values A. Push the "PAGE" button to select one of the four Memory Zones. B. Once one of the four Memory Zones has been selected, push the "MEMO" button twice. The first blood pressure reading will be displayed.
Clock Settings 1. To set time and date A. Press the " " button and hold for at least 5 seconds. You can start setting the year when the "YEAR" digit blinks. B. Then press " MEMO " to increase the year and "...
Note: If correct measurement is impossible after troubleshooting, consult store of purchase or Lumiscope repair department. (Do not disassemble unit). In some rare cases, there may be error due to the physical condition of the person. In such cases, please consult your physician. If the clock doesn't show on the LCD, please adjust the clock settings.
Lumiscope Repair Department. Service of this product by anyone other than the Lumiscope Repair Department voids warranty. In the event that Lumiscope does not abide by the terms of this warranty, the consumer may seek breach of contract remedies in the New York State Federal courts of law.