To avoid electric shock, do not perform any servicing other than that contained
in the Users or Service manual unless you are qualified to do so. These service
instructions are for use by qualified personnel only.
This Calibration Information for the 79/26 Series III Multimeter (hereafter known as "the Meter") provides
the information necessary to calibrate the meter and verify its performance. This information sheet provides
the following information:
Disassembly and reassembly
Performance test procedures
Calibration and calibration adjustment procedures
For complete operating instructions, refer to the 79/26 Series III Instruction Sheet.
The Meter has a limited lifetime warranty. To contact Fluke, call one of the following telephone numbers:
USA: 1-888-99-FLUKE (1-888-993-5853)
Canada: 1-800-36-FLUKE (1-800-363-5853)
Europe: +31 402-678-200
Anywhere in the world: +1-425-446-5500
For additional information about Fluke, its products, and services, visit Fluke's web site at:
PN 659349 December 1997 Rev.2, 3/00 ©1997, 1999, 2000 Fluke Corporation. All rights reserved. Printed in U.S.A.
Series III Multimeter