SEM Hitachi S-4700 user manual English Version by Eric, VDEC, Mita Lab, 2015/01/27 Please, leave this manual here. You can find an electronic version in the TakedaCR Wiki: http://takeda-cr.t.u-tokyo.ac.jp (accessible only from inside Utokyo) 1. Warnings and recommendations Be sure to read the following remarks: 1.1.
5.3.2. Fix tightly the sample on the sample holder with the double-sided conductive tape (don't use too much tape). If the sample drops into the chamber, the chamber will have to be disassembled, 5.3.3. Fix the height of the sample with the lock screw. The highest part of the sample should not exceed the level gauge by 0.5mm or it could damage the lens inside the main chamber.
7.3.2. Rotate the "STIGMA/ALIGNMENT" knob on the operation panel in order to make the displacement as small as possible. 7.3.3. Click "Stigma Alignment Y" and repeat the same operation. 7.3.4. Click "OFF" 7.4. Low Mag Position 7.4.1. In the mode "High Mag", set the lower magnification, and set a landmark on the sample in the center of the screen.
9.3.7. Wait for atmospheric pressure and pull the door open by grabbing the door itself, not the rod. 9.3.8. Push a little the rod and unscrew the specimen holder from the tip of the rod. 9.3.9. Pull the rod back into the locked position and close the door. 9.3.10.