©2016 - v 5/16
#631-0115 Advanced Microscope
MODEL SME-F6 Microscope, Monocular, Light. BIOLOGICAL MICROSCOPE
This information manual is intended to familiarize the first time user with their
instrument. These microscope are ruggedly built for classroom and laboratory
use. The instruments feature fully-coated achromatic optics and interna-
tional standard DIN objectives and eyepieces. Optically, they are excel-
lent systems with good resolution, centering, and paragon features.
Mechanical movements are smooth and positive.
Mechanical stage, and rack stop serve to minimize damage.
Magnification Vision field's
• Viewing Angle: Head inclined at 30°, 360°rotatable.
• Illumination: Built-in 220V/110V20W Incandescent lamp
(Max power: 20W), and there are an ON-OFF switch & a
fuse for prevention of short circuit. With focusing lens of
Abbe condenser (N.A 1.25) and iris diaphragm.
1. GETTING THE MICROSCOPE READY FOR USE
First, familiarize yourself both visually and physically with the mechanical parts of your microscope. Gently
operate each part by hand (no tools required) to see how it behaves and what result it produces.
2. MOUNTING THE OBJECTIVE AND EYEPIECE
To put the eyepiece in place simply slide it into the eyepiece tube. The objective lenses screw into the rotating
nosepiece. If the four objectives are already mounted in the nosepiece, check to see if they are tightly in place.
To do this, grip the knurled ring on the objectives with your fingers and turn it to the right. Do not tighten them
INFORMATION & INSTRUCTIONS
Please read the booklet before using the microscope.
| 86475 Gene Lasserre Blvd., Yulee, FL 32097 | 800-875-3214 | www.sciencefirst.com | firstname.lastname@example.org