7. Gyro Offset Adjustment
Adjustment location: EIS Adj of computer screen
Measuring location: Computer screen,
Note: Do not shake the camera while gyro offset is being
adjusted. There is no gyro gain adjustment.
1. It turns on power.
2. Connect the camera's USB terminal to the computer'
3. Double-click on the DscCalDi.exe.
4. Select "EIS Adj" on the LCD "Test".
5. Gyro offset adjustment value will appear on the screen.
6. Click the OK.
The adjustment values fulfill the conditions below, they are
determined as within specifications.
Adjustment value determination