Changing the Image Display Position (Partial Function)
ZFV-C allows you to increase the processing speed by narrowing the image acquisition
area. The image area can be moved, when measurement speed is FAST mode (1/2
screen) or MAX mode (1/4 screen).
Changing the Measurement Speed p.98
The center of the image is
selected just after the mea-
surement speed is
MENU mode-[IMAGE]-[DISP POS]
Move the image area using the UP/DOWN
Press the SET Key to fix the setting.
Increasing the Sensor Head Sensitivity
If the image brightness cannot be increased by either shutter speed or lighting setting,
increase the sensitivity magnification.
Setting value (magnification)
Setting Image Acquisition Conditions
Good (small amount of noise)
Bad (larger amount of noise)
Can be moved to the
desired position vertically.