6 • Service Menus
Allows you to re-zero the WOW manually for those instances where the sys-
tem cannot re-zero itself (e.g., after certain error conditions) or before running
the Static Weight procedure (see previous page).
This reports lists dates and times on which the WOW was calibrated.
This allows you to test the WOW load cell accuracy You place a 16 oz. test
weight on the WOW and the system takes five samples. The system lists
the average, expected, standard deviation, and "Pass/Fail" status results of
Last 10 Weights
This provides the weights of the last ten pieces processed by the WOW
This allows you to calibrate an attached platform scale by placing a known
test weight (4 lb.) on the scale.
Used for Brazil only.
DM300c/DM400c/DM475 Service Manual (SV61801 Rev. D)