1. Attach the shoulder strap to
the carrying case.
Slide SimplyGo into carrying case,
secure, and attach shoulder strap if
desired. SimplyGo can be operated in
or out of the case.
2. Insert the charged battery into
There is only one way the battery can be
inserted into SimplyGo. Using its handle,
position the battery over the battery
compartment, lower into place, and press
until it snaps into place. After the battery
is inserted, check to make sure the
battery handle is level with or slightly
below the top of the plastic case.
3. Connect a nasal cannula to the
oxygen outlet on top of the device.
Connect a nasal cannula to the top of
SimplyGo. To avoid a disruption of oxygen,
route the cannula so that it is not pinched
or kinked. A standard single lumen nasal
cannula and tubing should be used.
SimplyGo works with cannula tubing up to
30 ft. (9 m).