BiPAP AVAPS BiPAP S/T Quick start guide • Set up device • Start therapy • View information • Comfort settings • Undocking • Humidification • Humidifier settings...
Set up device To access provider settings: 1. From the main menu, highlight Setup. Therapy Comfort 2. Press and hold ramp button and control wheel simultaneously for 4-5 seconds until beep. Info Setup 3. Provider Setup screen then appears. To program device: Back 1.
Provider setup menu (cont.) Alarm settings Apnea 0, 10, 20, 30 sec Patient disconnect 0, 15, 60 sec Low min vent 0 (off) to 99 lpm in 1.0 increments Low Vte 0 (off) 1 (on) System settings Backlight On, Off Language English, Italian, Spanish, German, French, icons hPa/cm H...
Start therapy 1. From the main menu, turn the control wheel to Therapy Comfort Therapy and press. 2. The monitoring screen will be displayed and the Info Setup device will start. Note: You can select the measured parameter to display on the Monitor Pressure screen.
Comfort settings 1. From the patient screen, highlight Comfort. Therapy Comfort The Flex, Ramp, and Rise Time comfort features will be shown in this menu selection if enabled. Info Setup 2. Turn wheel to specific setting and press. 3. Turn control wheel to desired option and press to select. 4.
Humidification With device turned off, remove and fill chamber with water. 1. (a) Lift humidifier door until it locks in place. Do not force door higher than the locked position. (b) Grasp front of chamber and slide toward you. 2. (a) Gently push tab located inside round opening on top of chamber.
Humidifier settings LCD display screen Number settings Humidifier icon Control wheel/Push button 1. When blower is activated, the humidification icon and numerical setting will light up on the device. 2. Turn control wheel to adjust humidifier number to the desired setting (0 to 5) and press. The settings represent relative humidity levels that will remain consistent throughout the night even if room temperature changes.