# Bank Mobile; Constant Measurement Of "K" For Mtco; Constant Measurement Of "K" For Ktco - Siemens BTC-BR User Manual

Tachograph programmer and calibrator

BTC-BR User's Manual
2.4.5 – Roller Set for KTCO and MTCO
Steps
Contents of the display
1
1. AUTO MEAS W
2. MANUAL MEAS W
3. ROLLER SET
5
6
(
) – Scroll
/
ROLLER
2
MTCO
KTCO
5
6
(
/
) - Scroll
Note: To use other banks of test is necessary to calculate the "W" of equipment through the formula:
π
[(2
r)
8]
x
x
x
2.4.6 - Measurement of the Constant "K" for MTCO
Steps
Contents of the display
1
2. MANUAL MEAS W
3. ROLLER SET
4. CHECK K MEASUR ←
5
6
(
) – Sel.
/
CHECK K MEASURING
2
MTCO
KTCO
5
6
(
/
) - Scroll
CHECK K MEASURING
3
K = 08000 P/Km
2.4.7 – Measurement and adjustment of the constant "K" for KTCO
For the measurement of the constant "K" in a vehicle equipped with KTCO, just run the same previous
procedure from step 1, changing up the option in step 2 from MTCO to KTCO. In this case the readed
value can be adjusted through the programming keys and must be checked by repeating the previous
procedure.
ATTENTION: This function is not applicable to the mechanical tacrograph KTCO
Keys
E
(E) Enter
SET
E
(E) Enter
1000
x
Keys
E
(E) Exec.
E
(E) Enter
C
(C) Exit
1308.
Procedures
Select Roller Set
This function prepares the equipment for
taking measurements of the "W" factor using
the Portable Roll Test (1602.26.160.267F).
From this screen the instructions must be
followed from the manual of operation of the
Portable Roll Test.
where: π (pi) = 3,1416
r = radius of the roll of traction
Procedures
Select Check K Measuring
This function reads the value of the constant
Select MTCO
This function prepares the Calibrator to
inject a signal of speed and measure its
return, by calculating the value of "K".
The value presented is the pre-
programmed into the tachograph. In
case of MTCO, the value of "K" can
only
be
"PROGRAMMING".