Tab. 4-3: Keys of the operation panel
4 - 8
Used to change the frequency setting and
Press to display the following.
Displays the set frequency in the monitor mode
Currently set value is displayed during calibration
Displays the order in the faults history mode
RUN command for forward/reverse rotation.
The rotation direction can be selected by setting Pr. 40.
Used to stop RUN command.
Fault can be reset when protective function is activated (fault).
Used to change each setting mode.
Pressing PU/EXT simultaneously changes the operation mode
Mode switch over
(refer to section 4.3.3).
Pressing for a while (2s) can lock operation (refer to section
If pressed during operation, monitor changes as below:
Used to switch between the PU and external operation mode. When
using the external operation mode (operation using a separately
connected frequency setting potentiometer and start signal), press
this key to light up the EXT indication. (Press MODE simultanesouly
(0.5s) or change Pr. 79 setting to change to combined mode.)
PU: PU operation mode
EXT: External operation mode (Cancels PU stop also.)