Figure 1-19. Creating a User Flatness Array
For this example, refer to menu map 5, POWER.
The equipment setup shown in Figure 1-19 assumes that if your
setup has an external leveling configuration, the steps necessary
to correctly level have been followed. If you have questions about
external leveling refer to earlier paragraphs titled, "Externally
Leveling the Synthesizer ."
Zero and calibrate the power meter/sensor.
Connect the power sensor to the point where flatness corrected
power is desired.
Setup Synthesizer Parameters
On the synthesizer, press (PRESET).
P O W E R L E V E L
Create A Frequency List
On the synthesizer, press FREQUENCY (E).
Select List Menu Enter List Freq @ (GHz). This enters 5
automatically sets the offset to 0 dB and the dwell to 10 ms.
Enter 18, 13, 11, and 20 GHz to complete this example array.
Getting Started Advanced 1-37