The detector then automatically deactivates.
12. Bump Test Fail: If a previous bump test failed, the detector
beeps, vibrates, and BUMP FAIL displays:.
to acknowledge the alarm.
Bump test the detector before continuing operation.
For information regarding bump tests, refer to the MicroDock ll
If the detector passes the self-test, it enters normal operation. The LCD
displays the ambient gas reading.
The detector begins recording immediately. It records the
maximum (MAX) gas exposure,
the short-term exposure levels (STEL), and
calculates the time-weighted average (TWA).
If the detector fails the self-test, refer to Troubleshooting.