User Manual SICLOCK GPS1000
The absolute time precision for the individual DCF output interfaces is maximum 60 µs for the TTY interfaces and maximum 10
µs for the RS232 interface. The relative precision is better than 3 µs for both interfaces.
The relative precision specifies the maximum deviation of two sequential impulses. The absolute precision is the offset to the
atomic clock supplying the whole GPS system. If signals from only four satellites or less are received, a time interval of 20ms to
50ms resulting from the signal propagation delay from satellites to the earth, must be added to the absolute times and in this
case cannot be corrected by the receiver. For receiving signals from four or more satellites the inaccuracy of the receiver is maxi-
6.4. Antenna stand
Fig 11: Dimensioning Antenna Stand
The stand is fixed by 4 5mm diameter countersunk head screws.
Order numberr 2XV9450-1AR82-BA