Table 10 list some of the error messages the Meter may display and the conditions that may be causing the error.
Leads connected incorrectly.
Batteries low – function
Error: Date and Time need to be
Not enough memory for
Batteries critically low, replace
Table 10. Error Messages
Lead in A or mA/ A jack but
Leads in both A and mA/ A jacks.
Rotary switch set to measure current but no lead in either A or mA/ A jack.
Thermocouple wire is open or thermocouple junction is corroded.
No Thermocouple connected to the Meter's input.
The function selected requires a higher battery level to operate within specifications.
Batteries were left out too long and the Meter's date and time were lost.
When starting a recording session or saving screen data, the Meter does not have
enough memory to store the information.
Battery is too low to make measurements within stated specifications. The Meter will turn
off within 15 seconds after this message appears to preserve the Meter's date and time.
rotary switch not in corresponding A/mA or A