Setting the upper limit alarm temperature
An upper limit temperature alarm is provided with the Incubator. The alarm temperature can be changed
by using the following procedure.
1. In the Stand-by Setting Display, turn the upper limit regulator on the control panel using a small
screwdriver to set the desired upper limit alarm temperature. The upper limit temperature can be set
between 34 and 52
S t a n d - b y
T e m p e r a t u r e
C O 2
D e n s i t y
K e y
L o c k
H i g h
L i m i t
Set the upper limit alarm temperature (High Limit) to at least 5
Set the upper limit alarm temperature to at least 50
2. After setting the parameter, press the MENU Key to display the menu, select OK, and press the
ENTER Key. The alarm temperature will be saved.
Refer to the tables of alarms and safety functions on page 46 for details.
The alarm temperature will be changed at any time by turning the upper limit regulator even if the
Stand-by Setting Display is not displayed.
Set the upper limit alarm temperature after the chamber reaches the set temperature for operation.
S e t t i n g
3 7 . 0
( 0 . 0
5 . 0 %
( 0 . 0 %
( 0 . U n l o c k
5 2 . 0
C when H
M E N U
C - 5 0 . 0
2 0 . 0 % )
C a n c e l
1 . L o c k )
C higher than the chamber set
decontamination is performed.