# Correspondence Table Between Temperature And Temperature Data - Hitachi HIDIC EH-150 Instruction Manual

Resistance thermometer input module(eh-pt4)

7.4

## Correspondence Table between Temperature and Temperature Data

(1)Pt100/Pt1000 -50 to +400 ºC range
Table 7.4.1 Pt100 -50 to +400 ºC range Temperature conversion data
Temperature
Decimal
( ºC )
data
Note1)
-60
63078
-55
63283
-50
63488
-45
63693
-40
63898
-35
64102
-30
64307
-25
64512
-20
64717
-15
64922
-10
65126
-5
65331
0
0
5
205
10
410
15
614
20
819
25
1024
30
1229
35
1434
40
1638
45
1843
50
2048
55
2253
60
2458
65
2662
70
2867
75
3072
80
3277
85
3482
90
3686
95
3891
100
4096
Note 1) At the range from -50 to +400 ºC of Pt100, the input temperature range is from -50 to +400 ºC. But the
temperature data is output from -60 to +410 ºC.
Note 2) In case of Pt1000, the resistance is 10 times of PT100.
Pt100
data
resistance
( )
Note 2)
F666
72.33
F733
78.32
F800
80.31
F8CC
82.29
F999
84.27
FA66
86.25
FB33
88.22
FC00
90.19
FCCC
92.16
FD99
94.12
FE66
96.09
FF33
98.04
0000
100.00
00CC
101.95
0199
103.90
0266
105.85
0333
107.79
0400
109.73
04CC
111.67
0599
113.61
0666
115.54
0733
117.47
0800
119.40
08CC
121.32
0999
123.24
0A66
125.16
0B33
127.07
0C00
128.98
0CCC
130.89
0D99
132.80
0E66
134.70
0F33
136.60
1000
138.50
7.Collection of Temperature Data
Temperature
Decimal
( ºC )
data
Note1)
110
4506
120
4915
130
5325
140
5734
150
6144
160
6554
170
6963
180
7373
190
7782
200
8192
210
8602
220
9011
230
9421
240
9830
250
10240
260
10650
270
11059
280
11469
290
11878
300
12288
310
12698
320
13107
330
13517
340
13926
350
14336
360
14746
370
15155
380
15565
390
15974
400
16384
410
16794
7-5
Pt100
data
resistance
( )
Note 2)
1199
142.29
1333
146.06
14CC
149.82
1666
153.58
1800
157.31
1999
161.04
1B33
164.76
1CCC
168.46
1E66
172.16
2000
175.84
2199
179.51
2333
183.17
24CC
186.82
2666
190.45
2800
194.07
2999
197.69
2B33
201.29
2CCC
204.88
2E66
208.45
3000
212.02
3199
215.57
3333
219.12
34CC
222.65
3666
226.17
3800
229.67
3999
233.17
3B33
236.65
3CCC
240.13
3E66
243.59
4000
247.04
4199
250.48