ments On or
(If you do not assign a wave, and do not mark the position Blank, then
one will be assigned automatically when you plug in a transducer that
produces a wave.)
Step 4. Select the wave by pressing the TouchStrip.
If you want to exit editing without changing the wave, press the
Main Screen key.
You can also assign a wave to a position in the setup menu. Press the
Setup key then select Waves.
Changing the Waves Speed does not affect the speed of respiratory
waves. You must set the speed of the respiratory waves separately (see
"Changing the Speed of the Respiration Wave" on page 186).
To set the wave speeds for all the waves on the screen,
Step 1. Press the Setup key.
Step 2. Move the highlight to "Speed".
This will set the speed for the waves except the respiration waves.
Step 3. Press on the TouchStrip.
Step 4. Select the appropriate setting.
Step 5. Select "Resp Speed".
This will set the speed of the respiration waves.
Step 6. Select the appropriate setting.
Step 7. Exit the Setup menu.
To switch a particular measurement on or off, use the setup for that
measurement (see "Setting Up a Measurement" on page 41).
To check or change the alarm limits for a particular measurement, use
the setup for that measurement (see "Setting Up a Measurement" on