Installing Your Monitor
The M3046A Monitor can be used with an additional display as
required. The information displayed on the additional display is
identical to that on the monitor display.
The size and speed of the waveforms may differ from the
specification for the M3046A due to the different display size.
There are two displays available from Philips Medical Systems which
have been tested for use with the monitor. They can be ordered under
the following monitor option numbers:
M3080A #H65 15" Color Display (IEC 601-1 compliant, for use in
the patient vicinity).
M3080A #H71 21" Color Display. An isolation transformer must
be used with this display and the M1389A isolation transformer is
included in this option (IEC601-1-1 compliant, for use outside the
Other displays which meet the specifications for the VGA interface (see
"Monitor Performance Specifications" on page 402) may also be used.
These displays must either:
comply with IEC-601-1 (spilling proof, enclosure leakage current,
fixed power cable if isolation transformer used), when used in the
patient vicinity, or
comply with IEC 601-1-1 and be used with an isolation
transformer (e.g. M1389A), when used outside the patient
An additional display must be installed by a Philips Medical Systems
service engineer or authorized Service Representative. By the addition
of a display, the M3 monitor becomes a "system" and must be safety-
tested as such after installation. Detailed information about installation
and the required safety testing can be found in the Service Guide