Connecting the Measurement Server...
Server directly on
You can connect the Measurement Server to the monitor by mounting it
directly on the monitor:
Step 1. Make sure that your monitor has a link bar:
If your monitor does not have a link bar,
a. Position the link bar as shown in the diagram above.
Make sure that the guide is in the slot under the plug (which
connects to the Measurement Server).
b. Press the Link Bar into position, until it clicks.
c. Tighten the screw into the back of the monitor.
Step 2. Place the Measurement Server on the back of the monitor.
If it is not tight against the back of the monitor, slip it away from
the link bar until it is.
Step 3. Slip the Measurement Server forward until it clicks into place.
Installing Your Monitor