During this test, the monitor is not making any patient
measurements, measurement results are not collected, and patient
alarms are not active.
If you need to stop the test, select Test Signals in the Setup menu
a second time.
Step 1. Make sure that the Measurement Server is connected to the
monitor, and the monitor is switched on.
Step 2. Press the Setup key.
Step 3. Move the highlight to Test Signals.
Step 4. Press on the strip.
Check that you see:
Artificial ECG Wave.
Square wave, Pulse = 100, from 0 to 120mmHg
[from 0 to 15kPa]