Charging Indicator (
the detachable battery when it is being charged.
Number of Cycles Indication: Displays an indication
of the total number of cycles defined for the cough
treatment and the number of cycles remaining. This
replaces the time and date indication and is only
displayed when the active prescription mode is
Advanced Auto and therapy is being delivered.
Date and Time Indication: Displays the current date and
time. Located at the center of the top of the screen. In
Adv Auto mode, during therapy, date and time are not
When the device is in the Monitoring screen, displays the
pressure bar (Current Pressure) with pressure markings and
pressure units. When the device is not providing therapy,
this location is used to display messages and status.
Peak Cough Flow: Displays the most recent measured
peak cough flow. When the device is turned on, this
parameter displays dashes in Standby before the first
exhalation phase is completed. The Peak Cough Flow
value displayed during pre-therapy breaths corresponds
to the most recent peak expiratory flow measured during
: Displayed only when the Pulse Oximeter accessory
is connected to the device and the device is in Standby.
Displays the current SpO
the oximeter. If questionable or bad data is being
received from the oximeter, displays dashes.
): This indicator is placed over
reading being received from
Chapter 4 Viewing and Changing Settings