6. The device will automatically cycle between any pre-therapy breaths and the number of
coughs [Inhale (positive) to Exhale (negative) to Pause (atmospheric pressure)] for the desired
number of cycles, and end with a final post-therapy breath, if enabled.
7. After the therapy is complete, disconnect the patient from the device and clear secretions that
may have become visible in the mouth, throat, tracheostomy tube, or endotracheal tube.
8. Repeat as advised by your clinician.
3.5.4 Preset Settings
When setting the device up for the patient, the health care professional can define up to three
Preset settings. The health care professional can also limit the number of available presets for
use. Presets allow the user to quickly select a group of prescribed settings to provide therapy. See
Chapter 4 for more information on how to define the Preset settings.
CoughAssist T70 user manual