DreamStation CPAP Pro
DreamStation Auto CPAP
DreamStation BiPAP Pro
DreamStation Auto BiPAP
IMPORTANT! Remove this guide before giving the device to the patient. Only medical professionals should adjust pressure settings.
This guide provides you with instructions on how to access and navigate the provider screens used to modify device settings.
Refer to the User Manual for more information on using the DreamStation therapy devices.
The device is to be used only on the instruction of a licensed physician. The home care provider will make the correct
pressure settings and device configurations including accessories, according to the health care professional's prescription.
If you need to contact Philips Respironics directly, call the Philips Respironics Customer Service department at
1-724-387-4000 or +49 8152 93060.
Accessing the Provider Mode Screens
Accessing Provider mode unlocks settings that cannot be modified by the user. To access Provider mode:
1. Supply power to the device. First, plug the socket end of the AC power cord into the power supply. Then plug the
pronged end of the AC power cord into an electrical outlet that is not controlled by a wall switch. Finally, plug the power
supply cord's connector into the power inlet on the back of the device.
2. Once the device is powered, press and hold both the control dial and the Ramp button
3. You are now in Provider mode. You can choose between the following Provider mode screens.
on the device for at least 5
Patient Menu Return
- 1 -