When the device is in Limited Access mode, the following items appear on the Options screen:
1. From the Main Menu, use the Up/Down key to highlight the Data item.
2. Press the Right soft key (Select) to select the Data item.
The Data screen provides a summary of the last measured patient data and device settings, shown below. You
can use the Up/Down buttons to scroll through the information.
HR (Heart Rate)
Peak Cough Flow
SD Card Capacity
Detach Battery Serial Number
Detach Battery Cycles
Whenever an informational message occurs while the device is running, the
the screen. This indicates that an informational message is available and the user should check the
Information Log as soon as possible.
1. To access the Information Log, from the Main Menu, use the Up/Down key to highlight the Informa-
tion Log item.
2. Press the Right soft key (Select) to select the Information Log item.
The information log lists all messages that have been generated while the device is in use.
icon will display at the top of