4.4.4 Error Output
To select the desired output value that will be sent to the analog
output device upon an error, complete the following steps:
1. Press the
2. Press the
the Output menu.
3. Press the
enter the Error Output menu.
4. Use the
will appear next to the selected option to indicate that it is
activated. The following options are available, and you may
activate only one option at a time.
Only the first four options are displayed on the screen upon
entering this menu. When you scroll down to the fourth option
), a down arrow to the right of this option indicates that
an additional screen of options is available.
None (no error output is generated)
Low (an output of 4 mA is generated)
High (an output of 20 mA is generated)
Value (an error output at a programmable fixed value is
NAMUR (listed on second screen of options)
automatically deselect any previously selected output.
5. Press the
oxy.IQ User's Manual
key to enter the Main Menu.
key twice and then press the
key three times and then press the
keys to select the desired option and then
key to activate that error output. A check mark
key on a different error output will
key three times to return to measurement
Chapter 4. User Programming
key to enter