Loss of Power Alarm Test
1. While the device is providing therapy, remove the power connector and verify that Loss of
Power alarm sounds.
2. Reconnect power and verify that the device resumes providing therapy.
Important: When testing is complete, and before patient use, adjust the device to the appropriate patient settings.
Note: Refer to the User Manual for complete device specifications.