When the Therapy knob is moved to AED, AED View is displayed (see
Mode-related information includes:
AED Message Area: Displays important messages for the user while in AED Mode.
Shock Counter: Displays the number of shocks for the current event (including shocks delivered in
Manual Defibrillation Mode).
Delivered Energy: Displays the energy amount that was delivered for the most recent shock.
Replaces the Shock Counter for 15 seconds after the shock is delivered.
Selected Energy: Displays the configured energy for the current patient category.
Wave Sector 2: Displays the Capnogram or the SpO
combinations of options you have configured (see
Patient Contact Indicator: Graphical representation of the contact quality between the patient and
the multifunction electrode pads. Orange or red lights on the PCI indicate poor patient contact.
Adjust pads to optimize patient contact. Multiple green lights on the PCI indicate good contact is
established. The PCI is displayed when pads are attached and the device is charging or is charged. It
is not displayed when using paddles. When using paddles, use the PCI on the paddles as a patient
AED Pause/CPR Progress Bar: When in use, replaces the wave in Wave Sector 2 and tracks the
progress of the analysis pause and CPR periods.
Patient Category: Displays the current Patient Category. The patient category triggers specific alarm
limits and AED energy settings for defibrillation. The background color changes with the selected
patient category for Adult (orange) and Infant/Child (blue).
AED View Layout
NOTE: Only the ECG acquired through multifunction electrode pads is displayed in AED Mode.
Shock Counter/Delivered Energy
waveform depending upon which
5: AED Mode Option
on page 154) or the AED Pause/CPR