3: Operating the Device
The Status Area contains key device and patient information, see
Patient information includes:
Device information includes:
Clinical Alarm Area
Name/ID – If entered, displays the patient's name, patient's ID or nothing depending on
the current configuration. See
Date and Time – The current date and time
Paced Status – Internally paced or non-paced. The area remains blank if paced status has
not been selected. (This information is not displayed in Pacing and AED modes.)
Event Timer – The elapsed time for the current event, displayed in hours:minutes:seconds
Patient Category and Weight Range – Adult (defined as ≥ 25kg) or Infant/Child (defined as
< 25kg). The background color changes with the selected patient category for Adult
(orange) and Infant/Child (blue).
Clinical Alarm Indicator - Indicates the state of global alarms. If alarms are on and audio is
not paused, the indicator is blank. If they are off, the indicator says
Alarms Off icon
Technical Alarms - Technical alarms not related to the battery or pacing are displayed in the
Technical Alarms area. A triangle
on the display.
Battery Status – If a battery is installed, the battery icon indicates the level of charge
Battery Charge Level
* - See
"Low Battery Conditions"
Table 32 "General Settings"
indicates there are multiple alarms present, alternating
≥ 40%− < 65%
on page 27.
Technical Alarm Area
on page 149.
≥ 65%− < 90%
and uses the