Other consideration: The Efficia DFM100 is suitable for use in the presence of electrosurgery. Burn
hazard protection is provided via a 1K current-limiting resistor contained in each ECG lead wire.
Proper lead placement (see "Electrode Placement" on page 49) is important to reduce burn hazards
in the event of a defect in the electrosurgical equipment. Do not entangle the ECG cables with the
electrosurgical equipment wires; do not place the ECG cabling near the electrosurgical equipment's
Size: Approximately 7 in (17.8 cm) diagonal viewing area.
Type: Color TFT LCD.
Resolution: 800 x 480 pixels (VGA) with 32 brightness levels per color.
Sweep Speed: 25 mm/s ± 10% nominal (stationary trace; sweeping erase bar) for ECG and SpO
capnogram wave is 6.25 mm/s ± 10%.
Wave Viewing Time: 6.5 sec ± 10%.
Type: Rechargeable, Lithium Ion; See battery label for capacity information.
Approximate Dimensions: 28.5 mm (H) x 80 mm (W) x 145.7 mm (L); 1.1 in (H) x 3.1 in (W) x
5.7 in (L)
Approximate Weight: Approximately 0.44kg (1 lb)
Capacity: With a new fully charged battery, at 20 °C (68 °F), one of the following:
100 full-energy charge/shock cycles.
2.5 hours of monitoring (ECG, EtCO2 and SpO2 continuously monitored and NBP sampled
every 15 minutes) followed by 20 full-energy charge/shock cycles.
Two hours of pacing (180ppm at 140mA with 40msec pulse) and monitoring (ECG, EtCO2
and SpO2 continuously monitored and NBP sampled every 15 minutes).
Charge Time with Device Turned Off and AC Power Connected: With temperature at 25°C (77°
F), less than 3 hours to 100% capacity; less than 2 hours to 80% capacity.
Battery Indicators: Battery gauge on battery, capacity indicator on display, power indicators on front
of device; flashing RFU indicator, audio beep and
condition. When a low battery message first appears there is still enough energy for at least 10
minutes of monitoring and 6 maximum energy discharges.
messages on the display for low battery