Time from the initiation of analysis in AED mode until ready to deliver shock is less than 23 seconds
The device powers on in AED mode ready to deliver shock in less than:
Patient Impedance Range: Minimum: 25 ohm (external defibrillation); 15 ohm (internal
defibrillation); Maximum: 250 ohm. Actual functional range may exceed these values.
Manual Defibrillation Mode
Manual Output Energy (Selected): 1-10, 15, 20, 30, 50, 70, 100, 120, 150, 170, 200 Joules;
maximum energy limited to 50J with internal paddles.
Controls: On/Off Therapy knob, Charge, Shock, Sync, ECG Lead Select, Patient Selection, Print,
Mark Events, Reports, Alarms, Smart Select knob.
Energy Selection: Front panel Therapy knob.
Charge Control: Front panel button; button on external paddles.
Shock Control: Front panel button; buttons on external or switched internal paddles.
Synchronized Control: Front panel Sync button.
Synchronized Shock Timing: Maximum time from R-Wave detected to shock delivered is 25ms, as
measured with oscilloscope from peak of input QRS wave to leading edge of defibrillation discharge
into a 50 ohm test load.
Indicators: Text prompts, audio alerts, QRS beeper, battery status, Ready For Use (RFU), External
Power, Sync Mode.
Armed Indicators: Charging/charged tones, flashing shock button on front of panel and on external
paddles, energy level indicated on the display.
AC power only and at 90% of rated mains voltage.
A new, fully charged battery even after 15 discharges of maximum energy.
32 seconds with AC power only and at 90% of rated mains voltage.
24 seconds with a new, fully charged battery even after 15 discharges of maximum energy.
Smart Biphasic Waveform
Philips Smart Biphasic Waveform at 200J into 25-175 Ohms