This chapter includes:
Efficia DFM100 specifications. See below.
Symbol and abbreviation definitions, see Tables 60 and 61.
Electromagnetic Compatibility, see
Approximate Dimensions: 23.5 cm (H) x 29 cm (W) x 20.5 cm (D); 9.25 in (H) x 11.4 in (W) x 8
Approximate Weight (without battery): 5.66 kg; 12.5 lbs
Standard Operator Position: Within one meter (3 feet) of the device.
Power: Rechargeable Lithium Ion battery; AC power using a protectively grounded outlet.
Alarm Tone and Voice Message Volume Range: Maximum - 85 dB(A), Minimum - 45 dB(A).
Alarm Tone Volumes:
Imminent Shutdown - Continuous tone alternating between 1000 and 2100 Hz.
High Priority - Tone of 960 Hz lasting 0.5 sec repeated every second.
Medium Priority - Tone of 480 Hz lasting 1 sec repeated every two seconds.
Low Priority - Tone of 480 Hz lasting 0.25 sec repeated every two seconds.
Visual Alarm Characteristics:
High Priority - Flashing at 2 Hz with 50% duty cycle (a .25-sec flash twice every second).
Medium Priority - Flashing at 0.5 Hz with 50% duty cycle (a 1-sec flash every other second).
Low Priority - Constant on.
Waveform: Biphasic Truncated Exponential. Waveform parameters adjusted as a function of patient
Shock Delivery: Via multifunction electrode pads or paddles.
Shock Series: Configurable energy escalation in a series.
Leads Off Sensing and PCI Sensing for Pads/Paddles: Apply 500nA rms (571Hz); 200uA rms
on page 213.