Event Information (Continued)
Pacing stopped with reason
(Configured Event Text)
Logged when a a critical failure has occurred, preventing therapy delivery.
Logged when the algorithm begins analysis.
Logged when artifact has been detected.
Logged when the algorithm detects a shockable rhythm.
Logged when the algorithm detects a non-shockable rhythm.
Logged when the algorithm is unable to make a shock/no shock decision.
Logged when the device enters or exits a forced pause.
Logged when the device enters or exits a NSA pause.
Logged when the device enters or exits NSA monitoring.
Logged when the device enters or exits CPR pause.
Logged when pacing starts and when the mode is changed.
Logged when the Pacer Rate is changed.
Logged when the Pacer Output is changed.
Logged when pacing is paused.
Logged when pacing starts.
Logged when power is restored if pacing is interrupted or stopped due to a power
loss and the Therapy knob remains in the Pacer position.
Logged when the Pacer Output is less than the selected setting by 20 percent or
10mA (whichever is greater).
Logged when the device stops pacing. Reasons include:
: Logged when a pads off condition is detected.
pacing therapy is detected.
primary ECG lead for pacing.
Logged when the Mark Event button is pressed.
Logged when you select an entry from the Mark Event menu.
Logged when you hit the Print button on the front of the device.
Logged when there is a printer failure during an Operational Check.
: Logged when a device error that prevents delivery of
: Logged when the Therapy cable is disconnected from
: Logged when there is a leads-off condition with the
13: Data Management