13: Data Management
Mode and Energy Positions
Logged when device first turns on.
Logged when device is turned on after being turned off for less than 10 seconds.
Logged when the Therapy knob is turned to the Off position.
Logged when a critical device failure has been reported.
Logged when any out-of-range voltages are detected.
Logged when the device determines that one or more subsystems is not functioning
Logged when the device cannot determine when the last Autotest was run.
Logged when power was not available (for a week or longer) to run a scheduled
Logged during RFU testing if the battery can't sufficiently charge the device.
Logged at the start of an event and when mode or selected energy changes.
Logged when exiting a clinical mode.
Logged when patient category is set or changed.
Logged when internal paced status is set or changed.
Logged when battery is low.
Logged when you are pacing and the battery is low.
Logged when imminent shutdown warning is issued.
Logged when shutdown warning is issued.
Logged when the battery charge level is good.
Logged when the battery is not present.
Logged when communication with the battery has failed.
Logged when the battery has reached its end of life.
Logged when pads are applied to the patient.