About the Data Displayed
When trending is initially displayed, the most recent trending data appear in the far right
The display automatically updates as new data become available, with the newest data appearing
in the far right column of the display and older data moving over to the left. If the display is full,
the oldest displayed data is removed.
If you scroll horizontally to view older data, the Efficia DFM100 updates the newest data when
you scroll back.
If a parameter data point has invalid information, a
indicated by a question mark just before the numeric value. Unavailable data are indicated by a
If a parameter has not been measured during the display period, it is not listed in the far left
The Heart Rate parameter is always the top entry in the Trends Report. If your device has the
Trend Report. If your device has the EtCO
AwRR values are listed next. If your device has the NBP option and values for NBP are
available, they are listed after AwRR in the Trends Report.
For continuous trend data (HR, EtCO
the valid values during the time period.
NBP measurements are displayed with a timestamp.
The units of measure for trend data are not displayed in the Trend table and report.
Setting Trend Intervals
Trending data can be shown at selected intervals for up to the last 8 hours of monitoring. You can
adjust the display's time interval for the current incident to 1, 5, 10, 15, 30 or 60 minutes. The
default is 5 minutes.
To adjust the Trend Interval:
Confirm there is a Trends Report on the Efficia DFM100 screen.
Press the Smart Select knob.
Turn the Smart Select knob to highlight
Select the interval you want and press the knob.
Navigating Around the Trending Report
Use the horizontal scroll soft keys to scroll left and right (backward and forward in time) in the
Trending Report. The soft key is inactive (grayed out) if there is no more data to be viewed in that
If there are more lines of data than can be shown on the screen, turn the Smart Select knob to scroll
up and down the display.
NOTE: Make sure there are no active menus before using the Smart Select knob to scroll up and down the
Trends Report. If you do have an active menu, exit the menu before trying to scroll up or down in
the Trends Report.
Option and values for SpO
is displayed. Questionable data are
and Pulse are available, they are listed second and third in the
option and values are available, the EtCO
, pulse) the displayed number is the average of
and press the knob.