Pacing View appears when the Therapy knob is turned to the Pacer position. Pacing View includes a
status block which appears in Wave Sector 3 of the display (see
paced status is not displayed. Pacing-related information in Pacing View includes:
Pacing Markers: Markers, indicating a pace pulse was delivered, appear in Wave Sector 1 (and
in Wave Sector 2, if the wave is cascading) each time a pacer pulse is delivered.
R-Wave Arrows: R-Wave arrows appear in Wave Sector 1 (and in Wave Sector 2, if the wave is
cascading) when in Demand Mode pacing. R-Wave arrows do not appear on paced beats.
Pacing Status: Indicates the current pacing status.
Pacing Alarm: If there is a pacing-related alarm during pacing, the current pacing status is
replaced with an alarm message. See
Pacing Mode: Indicates if the device is in Demand or Fixed Mode Pacing.
Pacing Rate: Indicates the current pacing rate, including unit of measure.
Pacing Output: Indicates the current output, including unit of measure.
Alarms: Are on automatically.
Pacing View Layout
Pacing Control soft key: Press
to pause pacing; press
WARNING: If pacing is interrupted for any reason, you must press the
When pacing is active,
If the device is running on battery,
If pacing is not active,
to start or
is displayed when the device is on AC power.
Pacer Settings soft key: Press
bring up the menu for adjusting Pacer Rate and
53). The patient's internal
on page 97.
soft key to resume pacing.