3 Basic Operation
Discharge the patient.
Remove the transducer from the patient and, using a soft tissue, remove any gel from it. Then
clean the transducer.
Dock CL transducers to their base station so they can recharge.
Tear off the paper at the fold. To avoid misalignment of the recorder mechanism, NEVER pull on
the paper to advance it, or try to tear it other than at a fold (unless using the paper guide with the
Switch off the monitor.
Switching the Monitor to Standby
To switch the monitor to Standby:
Pressing any key or selecting any field on the screen will resume monitoring.
If an Avalon CL base station is connected to your FM20/30 monitor, do not turn off the monitor if
you need to recharge the batteries of the CL transducers. They can only be recharged if the FM20/30
monitor is on. If an Avalon CL base station is connected to an FM40/50 at the Telemetry ports in the
rear, the batteries of the CL transducers can be recharged while the monitor is in Standby mode. If you
disconnect a base station from a fetal monitor when the CL transducers are not fully charged, the base
station starts beeping. If you want to still disconnect it, confirm your choice by pressing the Standby
button of the base station, and the beeping stops.
menu using the SmartKey.