Deactivating Cableless Transducers
The CL transducers of the Avalon CL are deactivated by redocking them at a base station. To
deactivate all cableless transducers at once press the Standby button of the Avalon CL base station. See
the Avalon CTS Instructions for Use for the deactivation of the CTS transducers.
Unassigning Cableless Transducers
The CL transducers of the Avalon CL are unassigned by manually removing them in the
corresponding setup menu from the group of assigned cableless devices, or by unassigning them
directly in the
The fetal monitors FM20/FM30 and FM40/FM50 are compatible with the Avalon CL and
Avalon CTS Transducer Systems. Regard the following points for cableless monitoring:
You can connect one Avalon CL base station with a red connector to an FM20/FM30 or an
FM40/FM50 (fetal socket), or one Avalon CL base station with a black connector to an FM40/
FM50 (telemetry socket).
window. See the Avalon CTS Instructions for Use for the unassignment of the
3 Basic Operation