3 Basic Operation
What You See on the Monitor
When you connect a transducer or sensor, the measurement numeric appears on the screen.
Fetal heart rate measurements are labeled in the order in which you plug in the transducers for those
measurements. It does not matter which fetal sensor socket you use, as the monitor allocates a channel
automatically. For instance, when monitoring triplets, the first transducer you connect is automatically
allocated a channel, and the measurement is labeled
also chapters "Monitoring Twin FHRs" on page 159 and "Monitoring Triple FHRs" on page 167.
When you touch a measurement numeric on the screen, the setup menu for that measurement opens.
The fetal sensor socket to which the transducer for this measurement is connected is identified by the
transducer position indicator in the setup menu header.
Interface cable to Avalon CL and
Avalon CTS Cableless Fetal
Connect the black connector to one
of the two black sockets (marked
"Tele") on the rear of the monitor.
, the second
, and the third