3 Basic Operation
Connecting the Monitor to AC Mains
Always use the supplied power cord with the earthed mains plug to connect to an earthed AC
mains socket. Never adapt the mains plug from the fetal monitor to fit an unearthed AC mains
Check that the line frequency is correctly configured in the
FM20/FM30 only: The protective earth conductor is required for EMC purposes. It has no
protective function against electric shock. Double and/or reinforced insulation protects this
device against electric shock.
Do not use AC mains extension cords or multiple portable socket-outlets.
Always ensure that the monitor is positioned so that the AC mains plug is easily accessible, to allow
disconnection of the monitor from the AC mains.
Toco (M2734A) and Toco MP Transducer (M2734B)
Ultrasound Transducer (M2736A)
Transducer finder LED - lights up
on the transducer supporting to
identify the measurement source
"MP" for M2734B "Toco MP"
transducers (additionally capable
of providing the maternal pulse
Cable - connects to any of the
four fetal sensor sockets on the
The M2736AA US transducer is
identical to the M2736A US
transducer, including all specifications