Non Stress Test (NST) Analysis as Clinical Decision Support (CDS)
traces and to generate a printed NST report, equivalent to the NST report functionality in
OB TraceVue Rev. F and based on the NICHD guidelines from 1997.
The application can be adapted to domestic or hospital guidelines. A report of the NST analysis
can be printed automatically or on demand.
This software option is available for the whole product family, Avalon FM20 to FM50.
FHR Sound Source
A newly-connected fetal measurement (FHR or DFHR) automatically becomes the FHR sound
source, without the need for manual intervention. If you prefer the previous behavior, this can be
changed in Configuration Mode (
FHR Numeric Display
The FHR numeric display shows changes in fetal heart rate faster and is updated more frequently.
Improved FHR and DFHR Label Concept
A DFHR numeric (fetal heart rate from DECG measurement) now also has a unique number (
You can control fetal heart sound volume directly with the configurable SmartKeys (
Trace Separation On/Off Operation
Now you can switch the
To prevent the NBP measurement from being switched on or off accidentally, this operation is
now only available in Configuration Mode.
You can enter the NBP setup menu by touching the NBP numeric (on the screen) before the first
measurement is taken.
Algorithm enhancements (more tolerant of movement artifacts).
There is an individual Alarm Tone Volume setting for Yellow, Red and Cyan alarms.
The new alarm settings available in Configuration Mode are:
Trace Interpretation feature allows you to automatically interpret FHR
) to allow easier identification of the related trace in OB TraceVue/IntelliSpace
) on the monitor's screen.
SmartKey automatically turns into the
on or off in every FHR setup menu.
2 What's New
FHR Vol. Up
SmartKey when switching to