User Manual for the
WPS Series Wireless Pressure Sensor
Figure 5. Antenna Replacement
Table 8. Antenna Replacement Procedure
Honeywell recommends that the sensor be removed from service and moved to a clean area before servicing.
Remove the two screws holding the antenna radome to the sensor housing.
Unthread the antenna from the RP-SMA connector.
Inspect both antenna and sensor RP-SMA connectors for damage or debris, clean as needed.
Thread the new antenna's connector on to the antenna jack on the sensor housing.
Hand tighten antenna connector and tighten snugly with open-end wrench.
Caution! Do not overtighten antenna as it could twist in the housing and damage the antenna cable, or separate it
from the R.F. board.
Re-install antenna radome, fastening it with two screws, and tighten screws to 0,8 Nm [7.0 in-lb].
As required, new software may be uploaded over the air, into the sensor. This procedure may be performed while the sensor is in
service, and physically still connected to its process input. No disassembly of the sensor is required.
Software updating, if required, may be performed in the field, utilizing the
Wireless Device Manager User's Guide.
are downloadable from the relevant Honeywell support pages.
Honeywell Sensing and Control
Software updating will require image files for the specific part number of sensor device, and
These procedures are described in the