User Manual for the
WPS Series Wireless Pressure Sensor
The following replacement parts may be ordered from Honeywell Sensing & Control.
Table 6. Replacement Parts
The above batteries are also available from the
For other available antennas, antenna cables, and lightning arrestors, refer to the WPS Series Professional Install Guide.
6.2.1 When to Replace
When the sensor displays an E2 message or the
operation remaining before the batteries expire, unless the publication rate is operating at 1 update per second, then there is only one
week of operation remaining.
When batteries are removed or expired, all sensor configuration data, calibration data, and program data is retained in the sensor's
non-volatile memory (NVRAM).
Batteries may be replaced while the Sensor remains connected to the pressure being measured.
6.2.2 Battery Storage
Batteries should be kept in pairs, not mixed together with others from different vendors or of different shipments.
6.2.3. Transporting Batteries
When transporting or shipping Lithium Thionyl Chloride batteries, be aware that many regulations and restrictions apply. These bat-
teries are not permitted for transport aboard passenger aircraft. For shipping purposes, two "D" sized Lithium Thionyl Chloride cells
contain approximately 10 grams of lithium.
3.6V LITHIUM THIONYL CHLORIDE (Li-SOCI2) BATTERY, D size
Xeno company under part number XL-205F.
WDM Interface displays
a low battery warning message, there are 2-4 weeks of
Honeywell Sensing and Control